ASTM B117-2016标准来源：本站 发布时间：2018-10-31 16:26:00 阅读：293
本页面标准信息均来源于网络收集，或由参与标准制定的供应商提供，只作为参考使用，为确保测试结果的准确性、节省您的时间，如需ASTM B117-2016标准详细信息或ASTM B117-2016具体测试方法，与我们技术人员联系或给我们留言，我们将为您相关的技术支持。
1.1 This practice covers the apparatus, procedure, and conditions required to create and maintain the salt spray (fog) test environment. Suitable apparatus which may be used is described in Appendix X1.
1.2 This practice does not prescribe the type of test specimen or exposure periods to be used for a specific product, nor the interpretation to be given to the results.
1.3 The values stated in SI units are to be regarded as the standard. The values given in parentheses are for information only.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents
B368 Test Method for Copper-Accelerated Acetic Acid-Salt Spray (Fog) Testing (CASS Test)
D609 Practice for Preparation of Cold-Rolled Steel Panels for Testing Paint, Varnish, Conversion Coatings, and Related Coating Products
D1193 Specification for Reagent Water
D1654 Test Method for Evaluation of Painted or Coated Specimens Subjected to Corrosive Environments
E70 Test Method for pH of Aqueous Solutions With the Glass Electrode
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
G85 Practice for Modified Salt Spray (Fog) Testing
Q-Fog CRH盐雾试验箱用于ASTM B117-2016测试的步骤及设置